Material: Body-Liquid Crystal Polymer Cap- Liquid Crystal Polymer; Button- Liquid Crystal Polymer; Outlet- Polysulfone; Inlet Fitting- Plated Brass. Standard male oxygen DISS(swivel) The EMT Resuscitator may be attached to a portable pressure regulated oxygen cylinder or a central oxygen source. It may be used with a mask, endotracheal tube or tracheotomy tube.
Advanced pneumatic technology limits and regulates gas flow to 40 lpm during the manually triggered breath, but will allow flow rates of up to 115 lpm during a spontaneous breath. Addressing AHA guidelines, the EMT Resuscitator delivers a constant preset flow of 40 lpm, reducing the possibility of gastric insufflation during mask ventilation and improving the distribution of gas throughout the lung when ventilating through an endotracheal or tracheotomy tube. The EMT Resuscitator features built in antiasphyxiation ports that allow the patient to breathe outside mbient air in the event of source gas failure. In addition, a 360° swivel oxygen inlet attachment eliminates the problem of line twisting and tangling.
Operating Temperature: -10oF to 125oF
Filter: 25 micron, wire mesh
Outlet: 22mm outside diameter x 15mm inside diameter
Supply Pressure: 50+,-5 psig
Pressure to initiate a breath: -1 to -2.5cmH2O
Manually Triggered Flow: 40+,-5 LPM @50 psi supply
Demand Flow: 100LPM at 0 to -5cm H2O pressure
Pressure Limit: 60 +- 5cm H2O
Secondary Pressure Relief: 75 +- 5 cm H2O pressure Expiratory
Resistance: Less than 2cm H2O @ 100LPM flow
Work of Breathing: -5cm H2O pressure @ 115 LPM flow
Anti-Asphyxiation Entrainment: 55 liters per minute minimum @ -5cm H2O pressure